The compact DISOMAT® Tersus weighing terminal is ideal for use in a wide range of weighing applications, regardless of whether you’re focusing on operating. DISOMAT® Tersus to control a crane DISOMAT® Tersus panel-mounting unit in field housing DISOMAT® Tersus Crane Scale, see Manual BV-HAA. A Schenck model DISOMAT Tersus (VTG ) digital indicator as shown in between weighing ranges may be automatic or manual.
|Published (Last):||22 January 2014|
|PDF File Size:||6.70 Mb|
|ePub File Size:||15.50 Mb|
|Price:||Free* [*Free Regsitration Required]|
Secondary Displays Connectors 5. This means that the operation from the PC keyboard is exactly the same as on the device direct. For itemssee Fig.
Safety Instructions Use As Originally Intended The measuring system and its connected mechanical components are exclusively designed for weighing and controlling tasks. Remove separator plate, and the connector board is exposed. The presence of this symbol indicates that the machine or component used has the potential to cause damage to the disomxt system or other plant sections.
This value, too, can be used to check the wiring in case of functional faults. The ICP relay outputs can be loaded with max. Sources of Danger No danger originates from the measuring system itself, if it is properly installed and commissioned.
System Manual DISOMAT_百度文库
Plug fieldbus card complete with sub-connector from mounting kit onto the processor card component side pointing downward and affix using the three bolts.
The present manual is valid from firmware version Vxxyy The system identifies a missing memory module automatically and outputs an error message upon the next storage attempt. The ICP module interface parameters are fixed: This problem can be avoided with the use of a separate active terminal resistor, e. All fieldbus interfaces are designed as cards to be plugged into the tersu card. Filling or discharge scales can also be easily combined at the same time.
No special jumper position must be observed. In the safe area, the sealing ring may be exchanged; certified screwed connections must be replaced in set. The module is organized as ring buffer, i.
When using devices tested with low impact energy, ensure that inspection window is located in a protected environment. Disomah table below lists the available variants: Subject to change without prior notice. Further information about the cookies used.
Mechanical Dimensions Maximum mounting length on top hat rail: Fitting D is provided for fieldbus connectors. Parametrize etrsus as follows: Bolts can then be tightened easily with the use of a hex socket head spanner.
Remove clamping nut and press in plug flush with the screwed connection outer edge.
However, danger can arise from transit of material to be weighed e. For further reading For further information, please refer to the documents below.
For details, see manual BV-H Observe housing protection standards. If the appertaining load cells are also mounted in Zone 2, or Zone 22, protective circuitry to EExi can be ttersus.
Assemble parts of the dongle plastic housing using clips.
Legal-For-Trade Memory Modules VMM Designed for storage of legal-for-trade data, the legal-for-trade memory modules replace a tally-tape printer or any external legal-for-trade memory. A fieldbox with a high degree of protection is available. To ensure the high fisomat of protection: They are used with all industrial scales where the weight is to be read off from a distance.
Linearity Zero point stability Range stability?