ASTM F is a standard test method for shear testing calcium phosphate adhesion strength to metallic substrates under uniaxial shear stress. Calcium. The GF fixture meets the requirements of ASTM F for shear testing calcium phosphate and metallic coatings by the aligned interface method. ASTM F Standard Test Method for Shear Testing of Calcium Phosphate Coatings and Metallic Coatings.
|Published (Last):||14 December 2015|
|PDF File Size:||11.72 Mb|
|ePub File Size:||18.80 Mb|
|Price:||Free* [*Free Regsitration Required]|
Force range of 5 kN to kN 1, lbf tolbf Adjustable test space The most popular choice for static tension and compression tests These dual column testers are available in both tabletop and floor standing models.
ASTM F1044 Shear Testing for Calcium Phosphate and Metallic Coatings
Combining values from the two systems may result in non-conformance with the standard. Information developed using this test method may be useful for certain quality control and design purposes. Cure the adhesive agent and prepare a minimum of five test specimens. Consider the right grip for the job.
ASTM F Shear Fixture for Testing Calcium Phosphate and Metallic Coatings | GF
astj Apply the coating of interest to the face of each in the same manner as for actual production devices. Acquire a polymeric adhesive that is capable of bonding the coating on the test specimen with a shear strength of at least It assesses the degree of adhesion of coatings to substrates, or the internal cohesion of a coating in shear, parallel to the surface plane.
The values stated in each system may not be exact equivalents; therefore, each system shall be used independently of the other. Historical Version s – view previous versions of standard. Have a question about conducting the test for this standard?
ASTM F is a standard test method for shear testing calcium phosphate adhesion strength to metallic substrates under uniaxial shear stress. Misaligned grips can easily introduce bending stresses that are not included in the typical stress calculation.
Referenced Documents purchase separately The documents listed below are referenced within the subject standard but are not provided as sstm of the standard. Calcium phosphate and metallic coatings are used on a wide variety of metal substrates around the world, and commonly on orthopedic implants.
By means of our testing expertise and modular product design, we will help find the testing solution that is right for you. Link to Active This link will always route to the current Active version of the standard.
ASTM F Shear Testing for Calcium Phosphate and Metallic Coatings
We are confident that we have what you are looking for. It is the responsibility of the f104 of this standard to establish appropriate safety, health, and environmental practices and determine the applicability of regulatory limitations prior to use.
Give one of our application engineers a call today for help with creating the best budget and testing plan according to ASTM F We offer machines, apparatus, testers, tensile grips, test fixtures, clamps, holders, jigs, faces, jaws, extensometers, environmental and humidity chambers, furnaces, baths and custom solutions for unique applications. Record the f104 load applied.
Continue the test until the components have been completely separated. Testing Procedure Acquire a polymeric adhesive that is capable of bonding the coating on the test specimen with a shear strength of at least Next, apply a shear load to each test specimen at a constant rate of cross-head speed of 0. Measuring the adhesive quality of these products is essential due to their direct impact on implant degradation and bioactivity.
This newer version provides a means of testing and comparing different calcium phosphate thermal sprays to determine the high adhesion characteristics. The specimen being evaluated must be representative of the actual end-use coating.